August 2, 2012
Respironics, Inc., Trilogy 100, 200, and 202 Ventilators
The Missouri Department of Health and Senior Services received information regarding the recall Trilogy 100, 200, and 202 ventilators because of power supply components. If a ventilator's power supply fails, the ventilator may unexpectedly stop functioning, thereby stopping ventilation. The ventilator alarm may also fail to sound. Ventilator power supply failures can result in serious adverse health consequences, including death.
Model/Catalog Numbers: 1032800, 1040004, 1040007, 1054096, 1054260, CA1032800, R1040004, R1054655, and 1045151.
The ventilators were manufactured from January 1, 2012 through April 27, 2012 and distributed from March 1, 2012 through May 1, 2012.
The full recall can be found at www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm314226.htm.